EM IF-908 INTERFERENTIAL UNIT
Call us at 080-25353641 or
email us at email@example.com for pricing and other details.
The Interferential stimulator is a dual-channel electronic unit. It incorporates a digital display that indicates the operational modes and the output. It is equipped with an 8-bit microcomputer for controlling the system. The intensity, duration, frequency, and modulation of the electrical impulses can all be adjusted. The stimulator uses 4 electrodes producing deeper stimulation when compared to other types of electrical stimulation.
IF-908 is the interferential stimulator (IF) device which is a battery-operated device that includes two controllable output channels. Amplitude, mode, and duration of the electrical impulses can be altered with the buttons or knobs. The biggest difference between TENS and IF is the rate. The pulse rate of IF is around 4000-4100 Hz which is much higher than TENS (2-150 Hz). This high-frequency stimulation can easily pass through the skin and reduce pain more effectively.
IF-908 provides effective interferential stimulation treatment for the relief of chronic pain, especially for aching muscles of the back, arms, or legs. The controls are protected by a cover in order to prevent accidental changes to the settings. With 9 IF modes and premium quality adhesive electrodes, IF-908 provides you the highly effective nerve stimulation for pain relief and pain management.
|An adjustable 0-60mA peak into 500-ohm load each channel, constant current|
|10.7cm(L) x 6.2cm(W) x 2.7cm(H)|
|Approximately 150 grams (battery included)|
|4000Hz fixed (CH1)|
|Modulating Frequency||4004-4160Hz, Adjustable (CH2)|
IF Frequency Modes
|Constant Mode||Constant stimulation 4-160 bps, Adjustable|
|Auto Sweep||4-45 bps, 80-145 bps, 4 SET bps|
|Interference Pulse Frequency||4-160 bps, Adjustable, 4 bps/step|
|Output Configuration||Quad polar (4 electrodes)|
|Wave Form||Symmetrical Balanced Sine Wave|
|Sweep Time||15 seconds|
|Pulse Duration||125 us maximum|
|Patient Compliance Meter||Display total treatment time|
|Timer||15, 30 minute select able|
|Max Charge Per Pulse||15 Micro-coulombs Maximum|
|LCD||Displays mode, bps rate, abrupt/ramp, timer, and channels|
|Low Battery Indicator||A low battery indicator will show when the battery is low|
|Power Source||9V Battery/ Adapter|
|Tolerance||There may be 10% tolerance of settings and 20% tolerance of output|
|Lock||Prevents the patient from accidentally changing any of the set parameters|
|Frequency Shift||Five modes, 1/1 abrupt shift, 6/6 abrupt shift, 6/6 ramped, 10/10 abrupt shift, 10/10 ramped, bps can be adjusted|
|Frequency Shift Percent||Frequency shifts from 30% below the set frequency to 60% above and then return to 30% below the set frequency|